Buoy Type Oxygen Inhaler MSLFW01
Medical regulator is used for oxygen breathing in hospital.
1、 Mainly component name (Please see overleaf)
2、 Mainly function:
① Oxygen flow range: 1~ 10L/ min 1 〜15L/min
® Pressure reducing range: 2.0~3.0kg
@ Relief valve Pressure range: (3.5kg± 0.5kg)
3^ Appliance introduction:
(D Before using, please fill pour clear water into moisture bottle about one third reaching water line site.
(2) DF-2 blind nut straw into bolt head, make the pipe of flow meter square water line.
③ Put off regulator(9), put on oxygen tube switch, so you will see the data from needle position of pressure gauge(2). At last, straw flow on/off at anti-clockwise according to patients need, adjust switch.
®If patient need oxygen for a long time,cannse nose pipe to help patient absorb oxygen.
® Moisture bottle (20) circles at clockwise, adjust(17) oxygen hole to other site. 4、Attention point
① Before using the suction, you will check all parts bad or good and clean it by drying cloth.
- Pressure gauge(2), regulator(3) no oil, otherwise explode.
- Clearwater in moisture bottle keeps clean.
® There is adjustable nut(6) in the front of plunder shall be straw at clockwise.
⑤ Relief value(7) (15) keeps out of danger.
® Oxygen dosage is decided by doctor.
©After using suction, put off the oxygen tube switch, make rest oxygen out, pressure gage to zero site, last put off flow switch if using again, first put on the oxygen tube switch, avoid exploding.