Roll over image to zoom in
Click to open expanded view
Ventilator Machine MSLVM10
1.1 Main Performances
1.1.1 Basic Functions
——Sigh (deep breath);
1.1.2 Ventilation Modes
1.2 Technical Data
—Tidal volume range: no less than 50 to 1200ml, permissible deviation: ±20 %.
——Maximum minute ventilation: ≥ 18 L/min, permissible deviation: ±20 %.
——Oxygen concentration of output gas:21%~100%
——Ventilator compliance: ≤30 Ml/kPa
——Controlled ventilation (IPPV) frequency range: 0 ～ 99times/min, permissible deviation: ±15 %.
——I:E ratio: 4:1~1:4
——Maximum safety pressure: ≤6.0 KPa
——Oxygen consumption: the variation in the gas pressure in the cylinder should be less than or equal
to 1.5MPa/h when the ventilator operates on a 12250KPa / 40L medical oxygen cylinder continuously for one hour.
——Ptr： －0.4 ～ 1.0 KPa, permissible deviation: ±0.15 KPa
——Time to switch between controlled and assisted ventilation modes: 6s, permissible deviation: ＋1 s, －2 s.
——IMV frequency range: 1 ~ 12 times/min, permissible deviation: ±15%.
——PEEP range: no less than 0.1 ~ 1.0kPa.
——Sigh (deep breath): the inspiration time should be no less than 1.5 times of the original setting.
——Duration of end expiration plateau: 0.1~1.0s,
——Pressure limit range: 1.0~6.0kPa, permissible deviation: ±20 %
——The presentation of the spontaneous breathing frequency, overall respiratory rate and ventilation capacity
is refreshed once every minute.
——Continuous operation duration: the ventilator can operate continuously on a 24-hour basis on an AC utility main.
——Main unit net weight: 15kg, dimension (L*W*H): 390*320*310 (mm).